Free Express Shipping
A-scan is suitable for measuring the anterior chamber depth, lens thickness, vitreous body length, axial length and calculating the IOL power for an implanted lens.And also with the central area and the peripheral areas gain automatic compensation capability, accurate measurement of central and peripheral corneal thickness, is widely used in the preoperative examination and postoperative effect evaluation of refractive surgery.
A. Probe Frequency: 10 Mhz
B. Resolution: 0.01mm
C. Accuracy: ±0.02mm
D. Gain adjustable: 0-120dB
E. Measurement range: 15~35 mm (vel 1640m/s)
F. Measurements: ACD, Lens, Vitreous, and AXL
G. Examination modes: Normal, Aphakic, pseudophakic and dense cataract
H. Measurement mode: immersion and contact , auto and manual
I. IOL formulas: SRK-II,SRK-T,BINKHORST,HOLLADAY,HOFFER-Q,HAIGIS
1. Triangular model, portable and firm.
2. 5.7 “inch color LED touch screen.
3. Standard workstation software, easily for transfer and manage data.
4. Free software upgrade via USB interface.
5. Built-in thermal printer,information output conveniently.
6. Internal optional battery, practical for out-clinic diagnose.
Power Requirement : AC 100-240V 50-60Hz
Rated input: DC12V 3.0A
Dimension: 200*168*165(mm, L*W*H)
Weight: of main Unit approx: 1.7Kg (exclude the accessories)
Standard configuration :
10.0MHz A probe.
|4||Gage block for calibration.|
|5||Touch screen pen.|
Optional Configuration :
|4||Prager shell for Immersion|
5.7 “inch color LED touch screen.
15~35 mm (vel 1640m/s)
PICC Line Insertion, Thyroid, Breast, Vascular, Nerve, MSK (Musculoskeletal), IV intraveineous injection and plastic surgery, Joint Injections
>> Returns :
You have 7 calendar days to return an item from the date you received it.
To be eligible for a return, your item must be unused and in the same condition that you received it.
Your item must be in the original packaging.
Your item needs to have the receipt or proof of purchase.
Our Warranty covers defects in material, defects in workmanship under normal condition of use.
The product must be in its original packaging. Customer needs to ask for a Return Merchant Authorization (RMA) stating the exact technical issue and under which condition it did happen. Any documents or accessories that shipped with the product must be included in the package. Customer returns the package at his charge.
The company will repair the product, at no cost, once technical department confirms the defect happens under normal condition of use. The company will repair any broken parts of a product using new or replacement parts. The product can be exchanged with a new product. The company ships back the fixed or new item for Free, back to customer.
The Warranty Period for Physical Goods is 15 Months from the date of purchase.
> Warranty does not cover :
Damage of a product resulting from:
– Unauthorized modification of the product
– Natural disaster
– Theft or loss of the product